ROSEMOUNT Liquid Analytical Transmitter
ROSEMOUNT 1066 Liquid Analytical Transmitterr
Quick Star Guide:
1. For mechanical installation instructions, see page 14 for panel mounting and page 15 for pipe or wall mounting.
2. Wire the sensor to the main circuit board. See pages 21-23 for wiring instructions. Refer to the sensor instruction sheet for additional details. Make loop power connections
3. Once connections are secured and verified, apply DC power to the transmitter.
4. When the transmitter is powered up for the first time, Quick Start screens appear. Quick Start operating tips are as follows:
a. A highlighted field shows the position of the cursor.
b. To move the cursor left or right, use the keys to the left or right of the ENTER key. To scroll up or down or to increase or decrease the value of a digit use the keys above and below the ENTER key. Use the left or right keys to move the decimal point.
c. Press ENTER to store a setting. Press EXIT to leave without storing changes. Pressing EXIT during Quick Start returns the display to the initial start-up screen (select language).
If applicable, choose units of measurement.
For contacting and toroidal conductivity, choose the sensors type and enter the numeric cell constant using the keys.7. Choose temperature units: °C or °F.8. After the last step, the main display appears. The outputs are assigned to default values.